Voodoogor Up to tests can be programmed 6. Portable 65 Non Portable 2. Product List Supplier List. Suppliers with verified business licenses. Automatic Manual You could also discover the latest health, medical and pharmaceutical news updated daily sdmi-auto our trade resource center.
|Published (Last):||1 April 2005|
|PDF File Size:||5.10 Mb|
|ePub File Size:||10.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check the reagent instruction, or consult the reagent manufacturer or distributor to see if mixing of R1 and R2 are allowed prior to testing. Put the inlet tube into the calibrator to be tested and click Aspirate. If it is multiple-point calibration, you can proceed to the next step only when the calibration tests of all the calibrators are completed. Put the inlet tube into the sample blank, and click Aspirate. Hemolysis, icterus or lipemia in the samples may lead to unreliable test results, so a sample blank is recommended.
Put the inlet tube into the sample and click Aspirate. Repeat this to test other samples. Click QC and select controls from the drop-down list box C1-C3.
Put the inlet tube into the control to be tested and click Aspirate. Prepare proper amount of reagent according the number of the samples. If it is double-reagent analysis, it is recommended that R1 and R2 are mixed and then dispensed into respective tubes or other reaction containers.
Dispense calibrator into the tube with reagent in it. If K factor is manually entered, skip it Put the inlet tube into the sample blank, and click Aspirate. Put the inlet tube into the container where sample is mixed well with the reagent and click Aspirate. Put the inlet tube into the container where control is mixed well with the reagent and click Aspirate. For flow cell, if the system switches to other tests or after test of high concentration sample is run, wash the flow cell for cuvette, washing should be done after every test.
Otherwise, the result might be unreliable. The differences are: a During startup initialization, you do not have to wash the tubing b During test, you should open the optical window and remove the cuvette ; c During test the reactant is dispensed into the cuvette manually.
A cuvette cap specific for cuvette should be used to cover the cuvette see figure Press Aspirate button or Start to start the test. Patient information and test result can be edited to print the report. On the main screen, click Result Sample.
Select the sample to be edited and click Edit to enter the patient information including name, gender, age, etc. Click Print to print the patient report. Please follow the instructions displayed on the screen to wash the tubes with distilled water and then wait for 20 seconds. Shutdown the printer external printer and the analyzer. During this process, test information, system setup, patient information and test result will be saved.
BA 88A SEMI AUTO CHEMISTRY ANALYSER
2019 Mindray ba-88a semi-auto chemistry analyzer